The paper presents a method of measuring the angle of rotation and twist using a tilted fibre Bragg grating (TFBG) periodic structure with a tilt angle of 6◦, written into a single-mode optical fibre. It has been shown that the rotation of the sensor by 180◦ causes a change in the transmission coefficient from 0.5 to 0.84 at a wavelength of 1541.2 nm. As a result of measurements it was determined that the highest sensitivity can be obtained for angles from 30◦ to 70◦ in relation to the basic orientation. The change in the transmission spectrum occurs for cladding modes that change their intensity with the change in the polarization of light propagating through the grating. The same structure can also be used to measure the twist angle. The possibility of obtaining a TFBG twist by 200◦ over a length of 10 mm has been proved. This makes it possible to monitor both the angle of rotation and the twist of an optical fibre with the fabricated TFBG.
This article presents a linear strain measurement method insensitive to temperature variations and using fibre Bragg gratings. Two Bragg gratings were applied with periods selected to obtain partial coverage of their spectrum characteristics. One of the gratings was subjected to a tension strength. Placing both gratings in one thermal chamber allowed - through ensuring the same thermal conditions - to obtain insensitivity of the entire scheme to temperature variations. The gratings were recorded on the same optical fibre and reacted to temperature variations in the same degree. Value of strain was indicated based on the transmission spectrum characteristic of two grating schemes. The use of transmission, not reflectance, characteristics of the gratings allowed for a direct measurement of the spectrum, without having to use a coupler or optical circulators, and at the same time, this allowed to simplify the strain detection scheme. We proposed applying the continuous wavelet transform with characteristics of the spectrum scheme of two gratings for improvement of static properties. Especially, the thermal linearity and stability of the sensor was improved. For a strain range up to 750 μe, the non-linearity error of processing characteristic obtained was 0.45%. Thermal stability of the scheme proposed was defined as relative sensitivity of the power to temperature variations. The mean value of such relative sensitivity of the scheme proposed in the temperature scope of 21ºC-95ºC, amounted to 0.195.
Real-time monitoring of deformation of large structure parts is of great significance and the deformation of such structure parts is often accompanied with the change of curvature. The curvature can be obtained by measuring changes of strain, surface curve and modal displacement of the structure. However, many factors are faced with difficulty in measurement and low sensitivity at a small deformation level. In order to measure curvature in an effective way, a novel fibre Bragg grating (FBG) curvature sensor is proposed, which aims at removing the deficiencies of traditional methods in low precision and narrow adjusting. The sensor combines two FBGs with a specific structure of stainless steel elastomer. The elastomer can transfer the strain of the structure part to the FBG and then the FBG measures the strain to obtain the curvature. The performed simulation and experiment show that the sensor can effectively amplify the strain to the FBG through the unique structure of the elastomer, and the accuracy of the sensor used in the experiment is increased by 14% compared with that of the FBG used for direct measurement.
This paper presents a method of using a sensor with uniform Bragg grating with appropriately generated zone chirp. The presented method can be used for measuring two physical quantities, namely strain and temperature. By providing the same temperature sensitivity and different sensitivity to strain of two parts of a sensor, and experimental measurement of qualities of the proposed system and its calibration (experimental determination of sensitivity), verification of the results obtained from laboratory tests and the possibility of its practical implementation has been confirmed. The sensor grating was placed in such a way that its half was in the zone of a variable value of axial strain caused by changes of the cross-section of the sample. The other half, however, was in the zone of a constant cross-section of the sample and of constant value of strain, caused by the force stretching the sample. The obtained errors of non-linearity of processing characteristics for measuring strain and temperature of the proposed system were 2.7% and 1.5% respectively, while coefficients of sensitivity to strain and temperature were 0.77 x 10-6 m/e and 4.13 x 10-12 m/K respectively. The maximum differences between the values obtained from the indirect measurement and the set values were 110 μe for strain and 3.8°C for temperature, for a strain of 2500 μe and a temperature of 40°C.