In this paper, we present a fibre-optic sensor for simultaneous measurement of refractive index and thickness of liquid layers.We designed an experimental low-coherence setup with two broadband light sources and an extrinsic fibre-optic Fabry–Pérot interferometer acting as the sensing head.We examined how the refractive index of a liquid film and its thickness affect spectrum at the output of a fibre-optic interferometer. We performed a series of experiments using two light sources and only one sensing head. The spectra were collected in ranges of 1220#4;1340 nm and 1500#4;1640 nm. The obtained results show that using two spectra recorded simultaneously for two wavelength ranges enables to determine thickness in a range of 50#4;500 #22;m, and refractive index of a liquid film in a range of 1:00#4;1:41 RIU using only one sensing head.
Dynamic angle measurement (DAM) plays an important role in precision machining, aerospace, military and artificial intelligence. Because of its advantages including high sensitivity, solid state and miniaturization, fibre-optic gyroscope (FOG) has great application prospects in the field of DAM. In this paper, we propose a dynamic angle metrology method based on FOG and a rotary table to evaluate the DAMaccuracy with FOG. The system synchronously collects data from the FOG and rotary table, and analyses the DAM accuracy of the FOG for different sway conditions compared with that of the angle obtained from the rotary table. An angle encoder in the rotary table provides absolute or incremental angular displacement output with angular displacement measurement accuracy of 10′′ (0:0028◦) and angular displacement repeat positioning accuracy of 3′′ (0:00083◦), and can be used as an angle reference. The experimental results show that the DAM accuracy of the FOG is better than 0:0028◦ obtained with the angular encoder, and the absolute DAM accuracy of the FOG is better than 0:0048◦ for given conditions. At the same time, for the multi-path signal synchronization problem in the metrology field, this paper proposes a signal delay measurement method combining test and algorithm procedures, which can control a delay within 25 #22;s.
The phase jitter enables to assess quality of signals transmitted in a bi-directional, long-distance fibre optic link dedicated for dissemination of the time and frequency signals. In the paper, we are considering measurements of jitter using a phase detector the detected frequency signal and the reference signal are supplied to. To cover the wideband jitter spectrum the detected signal frequency is divided and – because of the aliasing process – higher spectral components are shifted down. We are also examining the influence of a residual jitter that occurs in the reference signal generated by filtering the jitter occurring in the same signal, whose phase fluctuations we intend to measure. Then, we are discussing the evaluation results, which were obtained by using the target fibre optic time and frequency transfer system.